- General Photograph
- Ray-filter Photograph
- Single bed and single bulb
- Design according to the electric mechanism of middle-size machine.The function is full and reasonable.
- The high-tension system is of march-past structure. Equipped with full-wave rectification high-voltage generator and independent X-ray tube unit, this machine is with large output power and operation is reliable and stable.
- Supply voltage V, roentgenoscopy, photography KV and step-less adjustable.
- Equipped with the manostat for filament of X-ray tube and the space-charge compensator as to ensure the stable and accurate output.
- Equipped with the capacity limitation and protection circuit and KvmAS interlock limitation and protection.
- Adopt the digital circuit timer, grade according to R10 priority coefficient and can control time exactly
- High-voltage primary uses the zero controlled circuit of silicon-controlled rectifier of large power.
- Photographic bed can move in length and breadth
- The photographic bed, pillar & vibrating ray-filter are in a whole without top & bottom track.
|Supply power||capacity||Two-phase 30KVA|
|voltage frequency||380V/220V ± 10% 50Hz ± 0.5Hz|
|inner resistance||0.75 Ω at 380V, 0.25 Ω at 220V|
|Power supply||Adjustable range||380V,380V ± 10% ,220V ,220V ± 10%，being adjustable continuously|
|Photography||voltage||50~100KV, be adjustable continuously|
|current||small focus 50mA,100mA , big focus 50,100,200,300mA|
|time||0.04~6.3s, 23 grades together according to R10 coefficient|
|Max. Dc output voltage||100KV|
|Max. Dc output current||200mA|
|focus||small focus 1X1mm,big focus 2X2mm|
|Photographic bed(YZ-I)||Bed face(LXWXH)||2000mmX665X700mm|
|Moving range of bed face||680mm in length,200mmin breadth|
|Ray-filter of bed||Travel in length ≧500mm|
|Grid density N28|
|Convergence distance fo100cm|
|Grid ratio R8|
|Pillar of X-ray source unit||Moving in length along photographic bed||1800mm or1600mm|
|Distance around as moving up and down(along upright post)||650-1800mm|
|Rotary around the center of cross arm||± 180°|
|Rotary around the shaft line of X-ray annular tubes||-10 ° ~ 60° ~ +120°|
|Max size of cassette for photograph||356mmX432mm(14”X17”)|