- urine based
- detects protein and creatinine
- results in 60 seconds
- surgery, clinic, or OTC use
- clear, precise instructions
- complements blood pressure and clinical history
- visual reading or by handheld instrument
- ideal as part of a maternity health programme
- OEM or branded label offered
Materials Required but not Supplied
Specimen collection container and timer
The sealed pouches in the test kit may be stored between 4-30°C for the duration of the shelf life as indicated on the pouch. When stored as indicated, the test kit will have a 24 month shelf life from date of manufacture.
- This kit is for in vitro diagnostic and professional use only.
- Read the instructions carefully before performing the test.
- This product does not contain any human source materials.
- Do not use kit contents after the expiry date.
- Handle all specimens as potentially infectious.
- Follow standard Lab procedure and biosafety guidelines for handling and disposal of potentially infective material. When the assay procedure is complete, dispose specimens after autoclaving them at 121ºC for at least 20 min. Alternatively, it can be treated with 0.5% Sodium Hypochlorite for 2 hours before disposal.
- Do not smoke or eat while performing assays.
- Wear gloves during the whole procedure.
Specimen Collection and preparation
- No prior special preparation of the patient is required before sample collection by approved techniques.
- The test works best on fresh whole blood samples. However, if testing cannot be done immediately, blood samples collected with a suitable anticoagulant such as EDTA or Heparin or Oxalate should be stored at 2-8ºC up to 3 days. Blood samples should not be frozen.
- Repeated freezing and thawing of the specimen should be avoided.
- Do not use haemolysed, clotted, contaminated, lipaemic and viscous/turbid specimen.
- Specimen containing precipitates or particulate matter must be centrifuged and the clear supernatant only used for testing.
- Do not heat inactivate the sample.
- Shipment of samples should comply with local regulations for transport of etiologic agents.
- Bring the kit components to room temperature before testing.
- Open the foil pouch and remove the test. Once opened, the test must be used immediately.
- Label the test with patient’s identity (name or number).
- Use the pipette provided with the test (inside the foil pouch) to add 2 drops (80 μL) of whole blood, serum or plasma to the sample well marked “S”.
- At the end of 15 minutes read the results.
Note: Some positive results may appear as early as 10 minutes.
Two coloured bands appear within 15 minutes. One coloured band appears in the Control Zone (C) and another coloured band appears in the Test Zone (T). The test result is positive and valid. No matter how faint the coloured band appears in the Test Zone (T), the test result should be considered as positive result.
One coloured band appears in the Control Zone (C) within 15 minutes. No coloured band appears in the Test Zone (T). The test result is negative and valid.
No coloured band appears in the Control Zone (C) within 15 minutes. The test result is invalid. Repeat the test with a new test device.
NOTE: After 15 minutes the result is valid for up to 1 hour.
|ELISA Syphilis Test|
In comparison with FTA-ABS performed on 459 samples (80 positive and 379 negative), the Test-it Syphilis Whole Blood Test showed a sensitivity of 97.5% and specificity of 99.5%.
No interference was found with bilirubin (10 mg/dl), haemoglobin (20 mg/dl) or triglycerides (600 mg/dl) on the sensitivity and specificity of the test.
- The test is for the qualitative detection of anti-Treponema pallidum antibodies in human serum, plasma or whole blood. It does not indicate the quantity of the antibodies or degree of infection.
- The test is for in vitro diagnostic use only.
- As in case of all diagnostic tests, a definitive clinical diagnosis should not be based on the result of a single test but should rather be made after all the clinical findings have been evaluated.
Holmes KK, Lemon SM, Mardh P, Piot P, Sparling PF, Stamm WE, Wasserheit JM, Weisner PF. Chapters 33-36. In Sexually transmitted diseases, 3rd ed. New York: McGraw-Hill,1999.[/bsf-info-box]
The tests can be performed using whole blood or serum.
What sample preparation is recommended?
Serum should be prepared in the same way as routinely performed for any serological assay. Freshly collected samples should be used. Serum samples stored at -20 °C may be used as well. Venipuncture whole blood samples may be used provided the blood has been collected with an anticoagulant such as EDTA, heparin or citrate. Alternatively for a finger prick sample, clean the patient’s finger with the alcohol swab making sure the alcohol is dry before pricking with the lancet provided.
How long will the shipment take?
Depending on the size and availability of the order, anywhere from 1-4 weeks. Please contact the manufacturer to confirm.
When does the product expire?
The product expires two years after the date of manufacture.[/info_list_item][info_list_item list_title=”Manufacturer’s Warranty: ” list_icon=”Defaults-thumbs-o-up”]Product replacements on faulty/non-performing product as per Life Assay’s quality control check procedures on retainer samples.[/info_list_item][/info_list]
Request More Information